Modified on: Tue, 24 Jul, 2018 at 9:44 PM
Is it necessary to calibrate the probes during their usage?
- No. The probes are calibrated in our production and should carry properties forever.
What is the accuracy of Powersense DC?
- The declared precision is up to 2% from the nominal current (we offer 50A, 100A or 400A transducers), so for the 400A the effective precision is +-8A! So to measure low currents, with 400A is not a good set-up. Try to make the best estimation for your actual needs and choose the suitable transducers.
Did you find it helpful?
Sorry we couldn't be helpful. Help us improve this article with your feedback.